It is fine to squirt Infacol down the biopsy channel in a syringe but it should not be added to the water bottle. The biopsy channel is flushed following the procedure by the bedside clean; this is a working channel and will not cause blockages in the scope.
Articles in this section
- The decontamination of endoscopes for my service is undertaken off-site. Do I still need to provide an IHEEM report?
- My decontamination facility has just opened - do we need to provide an IHEEM report?
- Should our decontamination lead be accountable to the theatre manager or an executive board member?
- Can my unit be accredited if an external person supports our decontamination team?
- Can Infacol be used through the biopsy channel in an Endogator pump?
- Does JAG have any guidance on centralised decontamination?
- If decontamination equipment becomes out of service for a number of days, do the centres where endoscopes are sent for reprocessing need to be JAG accredited?
- Our plan is to move to a pass through washer and the vacuum pack system, is this sufficient to fulfil JAG criteria or do we still need a drying cabinet?
- Should the 3 hour rule be taken from the disinfection time of the scope or from the time the scope is removed from the AER?
- Can Tristel wipes be used to clean nasoendoscopes?