It is fine to squirt Infacol down the biopsy channel in a syringe but it should not be added to the water bottle. The biopsy channel is flushed following the procedure by the bedside clean; this is a working channel and will not cause blockages in the scope.
Articles in this section
- Can my unit be accredited if an external person supports our decontamination team?
- Is it okay to use Infacol through the biopsy channel in an Endogator pump?
- Does JAG have any guidance on centralised decontamination?
- In the event of decontamination equipment being out of service for a number of days. Do the centres where endoscopes are sent for reprocessing need to be JAG Accredited?
- Our plan is to move to a pass through washer and the vacuum pack system, is this sufficient to fulfil JAG criteria or do we still need a drying cabinet?
- Should the 3 hour rule be taken from the disinfection time of the scope or from the time the scope is removed from the AER?
- Can Tristel wipes be used to clean nasoendoscopes?
- We have hard containers for moving the scopes, must they be locked whilst they are being moved?
- Does JAG accredit services which provide outsourced decontamination of endoscopes?
- What are the decontamination standards for JAG?